Nickel Chrome Dental Casting Alloys for Porcelain and Ceramic in
Ingot or Cylinder Shape
Our Ingot Shape of Nickel Chrome dental alloy is a Nickel-based
alloy that possess an excellent workability and absolutely
versatile characteristics. It compates with most ceramics powders
seamlessly. Thanks to Willett's patented special formula, it
consisted superior bonding strength compares to other competitors.
|Melting range ℃||Density g/cc||Yield Strength(Rp0.2)|
1. Extremely high resistance to corrosion: no cytotoxic potential
in cell culture test
2. Biocompatibility certified by Both government and third-party
3. Low thermal conductivity
4. High modulus of elasticity
5. Great strength with any acceptable span
6. Long cooling time not required: low coefficient of expansion
7. Reliable metal-to-ceramic bond
8. High heat resistance: consistence of dimensional stability
during high-temp pocessing
*Customers can choose with NCP engraved or without.
What is the perfect temperature to obtain the best result when
The casting temperature should always maintain 100~150℃ higher than
What is the perfect timing to obtain the best result when casting?
While casting Nickel Chrome alloys that don't contain Beryllium,
the surface of the melting alloys will appear a layer of shadow.
Once the shadow disappears and the edges of the alloy have become
rounded (melted until it becomes a sphere shape is not necessary)
will be the best timing for casting. Perfect timing for casting
Cobalt Chrome alloy is to heat it till the surface membrane
perforated and a spark of bright light start to appear.
Cobalt Chrome casting piece is incomplete, what are the causes?
1. Casting ring is not heated evenly: While casting Cobalt alloys,
casting ring must heat up gradually and slowly before reaches 300℃.
Let it sit in the oven in between 300℃ and 600℃ for 10-30 minutes.
Once it heats up to the desired temperature, maintain the same
temperature for at least 30 minutes before casting.
2. Casting temperature is below/above alloy's requirement.
3. Strub designed improperly: For example, Strub is too narrow. We
strongly recommend the Strub diameter consist of 5mm or more.
4. The amount of alloy is not enough for the casting piece or the
centrifugal force of casting pressure is not enough.
5. Casting piece has not avoided the heat center.